Microsoft word - journal of advancement in medicine.doc
Journal of Advancement in Medicine Volume 8, Number 2, Summer 1995 The Basis for Microcurrent Electrical Therapy in Conventional Medical Practice Joseph M. Mercola, DO and Daniel L Kirsch, PhD, DAAPM ABSTRACT:
The use of electricity in medicine is not new. Clinicians used it over 150 years ago to treat non-union bone fractures. Electomedicine and nutrition, abandoned early in this century, have been recently revived. Most physicians are unaware of their therapeutic benefits. Electrotherapy, especiaily microcurrent electrical therapy (MET) is usefil for a variety of clinical conditions. Indeed, it may be the best treatment for many pain-related disorders, providing fast relief of symptoms and quickly promoting healing. It has significantly less side effects than drugs in chronic conditions. The more advanced MET devices can often demonstrate effectiveness with a simple two minute office procedure, allowing validity to be quickly assessed.
Pain is a serious problem that only recently has been getting the attention that it deserves. It and its associated symptoms have a potent economic impact. The Interagency Committee of New Therapies for Pain and Discomfort estimates that chronic pain affects more than 40 million Americans and costs the US economy over $65-$70 billion annually. At least 10% of Americans suffer chronic, handicapping pain. The avenge chronic pain patient has suffered for seven years and has had 3 to 5 surgical operations, spending $50,000 to $100,000 or more. lost productivity due to pain is estimated at over 700 million work-days per year (l).
Although pain may be an important warning of a disease process, it often has limited diagnostic value and remains a difficult problem for the physician. A recent study (2) examined visits to eclectic and alternative medicine practitioners. It reported that non-reimbursable costs were about $10.3 billion in one year, comparable to the $12.8 billion hospital expenses during the same time period. In 1990, Americans made an estimated 425 million visits to these eclectic practitioners. while making only 388 million visits to all US primary care physicians. Many patients cite the side-effects and short-term relief of drug therapy as the primary reason that they seek alternative medical care. New development in electromedical technology offers physicians an effective treatment for pain-related disorders for many of them.
Traditional Therapy and TENS
Electrical modalities have been ueed for many years to control both acute and chronic pain. Clinicians also routinely use neuromuscular electrical stimulators to rehabilitate injured athletes (3,4). Transcutaneous electrical nerve stimulation (TENS) and other
similar devices use a mild form of electrically induced pain to block the body's ability to perceive the pain that is being treated (5,6). when patients receive TENS at unmasked low frequencies (eight pulses per second or less) their production of endorphins may increase, thus producing temporary relief, possible in approximately 50 per cent of people. The effect of TENS is believed to stimulate A-beta pain-suppressing nerve fibers to overwhelm chronic pain-carrying C fibers (7). Similar results can be achieved by repeatedly tapping the painful areas with a blunt object. Massage, ice and heat relieve pain this way. The ampere (amp) is the measure of electron movement or current past a fixed point over time. Interferential, TENS, and high-voltage pulsed galvanic stimulators deliver currents in the milliamp range, stimulation which generally exceeds nerve firing thresholds, resulting in sensation ranging from a gentle tingling to intense muscle throbbing.
Traditional TENS only works if the current is strong enough to feel, using a current up to 80 milliamps. Patients are advised to set the current at the maximum comfortable tolerance, but the nervous system gradually accommodates to this high level of current, causing tolerance similar to that of chemical analgesics. Increasing the current causes mild electrical burns in about one third of the patients. The technique provides no significant residual effect.
Microcurrent Electrical Stimulation (MET)
Microcurrent electrical therapy represents a significant improvement in rapid pain control and acceleration of healing. It uses current in the microampere range, 1000 times less than that of TENS and below sensation threshold. The pulse width, or length of time that the current is delivered with a microcurrent device is much longer than previous technologies. A typical microcurrent pulse is about 0.5 seconds, which is 2500 times longer than the pulse in a typical TENS unit and a good microcurrent unit has approximately ten times the electronic circuitry of a TENS unit.
Unlike TENS, MET is usually administered through hand held probes positioned so that current flows between them, through the painful area, for ten seconds. The vast majority of pain problems can be treated with less than 10 applications of 10 second probe treatments. Many patients are free of their pain in less than two minutes and there is generally a significant residual effect, often lasting from at least 8 hours to as long as 3 weeks or more (8). The first homecare MET stimulator was introduced in 1982. It provides at least the same results as more expensive models (9). It is a pocket-size device for home use and patients find it easy to learn to use it, as necessary, to control their pain.
How Microcurrent Works
MET works because of its ability to stimulate cellular physiology and growth. One classic study (10) showed that it could increase ATP generation by almost 500%. Increasing current actually decreased the results. This study also demonstrated its ability to enhance amino acid transport and protein synthesis. One can see an illustration of the true therapeutic effect of MET through the mechanism in which trauma affects the
electrical potential of damaged cells (11). The injured area has a higher electrical resistance than the Surrounding tissue. This results in decreased electrical conductance through the injured area and decreased cellular capacitance (12), leading to impairment of the healing process and inflammation.
Correct application of MET to an injured site augments the endogenous current flow, allowing cells in the traumatized area to regain their capacitance. Resistance is reduced, allowing bioelectricity to flow through and reestablish homeostasis. This process helps to initiate and perpetuate the many biochemical reactions that occur in healing. Muscular spasm, occurring as a reaction to trauma, causes reduction in blood supply, resulting in local hypoxia, accumulation of noxious metabolites, and pain. This, in turn, leads to reduction of ATP synthesis. Thus, MET stimulation results in replenishment of ATP (10).
Rapid Pain Management
One of the greatest values of MET is in pain control (8,9,12). It also reduces inflammation, edema and swelling, increases range of motion, strength, and muscle relaxation, and accelerates wound healing (13,14). It is exceptionally useful in soft tissue injuries, such as sprains (15,16). wounds, post-surgical trauma, and particularly in treatment of long-term residual pain due to post-surgical scars. It is effective for treatment of headaches, temporomandibular joint syndrome, neuropathies, arthritis, bursitis and tendonitis. Clinical experience indicates that it is an adjunctive therapy in earaches, sore throats, toothache, sinus congestion, viral or allergic conjunctivitis post-herpetic neuralgia, skin ulcers, post-CVA spasticity, and compression neuropathies such as carpal tunnel syndrome. It has also proven useful in preventing the delayed muscle soreness that is common after heavy exercise (17). Improvement in post-exercise muscle fatigue was achieved by applying the current over the exercise muscles for twenty minutes after exercise. In a minority of patients MET does not work or only provides brief palliative relief. Its full potential is yet to be defined.
It has been used to control hypertension (18), failed back syndrome (19,20). arthritis (21), Raynaud's phenomenon (22,23), tinnitus (24-26), and post-anesthesia emesis (27). Dentists have used it as a substitute for local anesthesia (28,29) and to control pain associated with orthodontic treatment (30).
Intractable pain in patients with head and neck cancer has been successfully treated with MET, even in some cases that were morphine resistant (8,12). After only 10 minutes of MET, pain relief lasted from 8 hours to more than 3 weeks. The technique has been used successfully at the University of Texas MD Anderson Center (31).
About 5% of long-bone fractures in the United States result in non-union (82). Electrical stimulation of the fracture provides a non-surgical option for repair. It is also being investigated for use in osteonecrosis and osteoporosis (83).
Using electrical therapy to heal non-union fractures is not new. It was first reported over 150 years ago (34,35). At the turn of the century, however, a number of medical charlatans, using electrotherapy, forced the Carnegie Foundation to have the Flemer commission review its use. In 1910, the Fleiner Report relegated electrotherapy to a scientifically unsupportable position, causing it to fade from medical practice. Further exploration of the technique was reported by Yasuda and Fukuda (86) who found that mechanically stressed bone produces a small negative electrical direct current that stimulates bane prodution.
Becker (37) performed research that led to applying electrotherapy to the healing of bone fractures (38). By 1976, over 100 articles had been published describing the effects of electricity on bone growth and repair in laboratory animals and in humans (89). As of 1990, more than 100,000 cases of non-union fractures and aseptic necrosis have been succesefully treated with electrotherapy (40).
Several methods are available to stimulate bone growth. All require 3 to 6 months of treatment, and have similar contraindications. A gap in the fracture greater than half the diameter of the bone or synovial pseudoarthrosis will result in failure (33).
The first clinical trial of direct current surgical implant in humans in the United States (41) achieved results in 4 months in a large percentage of cases (42). Stainless steel electrodes with 5-20 microamps of current produced the best growth, while current above twenty microamps actually caused hone to die (43).
A noninvasive alternative is inductive stimulation, which works by creating a magnetic field around the non-union site. Pulsing electromagnetic fields (PEMFs) are induced by a treatment coil or transducer. These devices are battery powered and portable. Patients wear them for 3 to 10 hours a day and treatment lasts about 6 months. Many investigators report 90% healing rates with this method (44). Although PEMFs contain both electrical and magnetic fields, the bone remodeling processes appear to respond mostly to the electrical field component. The magnetic field contributes less benefit to the process (45).
Spectral analysis of PEMF frequencies shows that they range from 1-250,000 Hz. As indicated above, the electrical, not the magnetic energy, is responsible for producing bone growth. Investigators tested 150, 75, and 15 Hz sinusoidal electrical field effects on the prevention of osteoporosis (46). They found that the 150 Hz field did not increase bone mass, but inhibited normal bone loss associated with disuse. The 75 Hz field increased bone mass by 5%. while the 15 Hz field actually increased it by 20%. The energy represented by this frequency is less than 0.1% of the PEMF field. This strongly suggests that the vast majority of the energy introduced by PEMF has no beneficial effects on bone regrowth and it is also probable that even lower frequencies, like the 0.5 Hz field produced by MET would provide even more impressive results.
Several devices use capacitive-coupled stimulation which produces an electrical field at the fracture site. They are 9 volt battery units attached to the skin over the fracture site. It has the advantage of not requiring precise placement of the electrodes and can be administered 24 hours a day. Unlike inductive coupling, patients using this treatment can have a full weigh-bearing cast and this tremendously enhances patient compliance.
The first capacitive-coupling devices used a 60 KHz sinusoidal wave form and delivered a current of 7 to 10 milliamps (47,48),but subsequent work suggested that non-sinusoidal wave form and much less current is more effective in promoting bone healing (10,11,49) Although clinical experience exists, no studies have been published to date for these applications with MET.
Tendon and Ligament Repair
One of the first studies published on treatment of soft tissue injuries was by Wilson in 1972 (50). Microcurrent delivered in a PEMF format has been helpful in the management of refractoty tendonitis of the shoulder (51). Stanish (15,16) used implantable electrodes with constant 20 microamp direct current in severed dog tendons. He observed a 92% return to normal in 8 weeks, compared to 50% in control animals.
Although implantable electrodes were used, it is likely that external electrodes could produce similar results. This could significantly enhance the current treatment of tendon ruptures. Use of MET seems to enhance cell multiplication in connective tissue, and speeds formation of new collagen in injured tendons. Accelerated healing of ligament and tendon injuries has been reported (52) and it has been shown to increase rat tendon healing by over 250%.
Chronic wounds, of which leg ulcerations make up a major share, are a therapeutic problem. It is estimated that 90% of leg ulcers are due to venous stasis, affecting 0.6 of men and 2.1% of women in their 60s (40,53). Acute soft tissue injury is common and there are 2.5 million burn wounds a year in the US. Of 30 million lacerations, one in 5 are serious enough to require auxiliary treatment (14). Use of MET is simple, safe, and efficient and can have tremendous influence on improving wound healing.
Becker (54) showed that living tissues have multiple direct current surface potentials which are combined to form a steady state bio-electric field. He hypothesized that injury causes a localized shift in the current flow, triggering repair. He called this the current of injury (COI). Although first described hy Galvani in 1786, and later by others (14) COI was finally confirmed in 1980 (55). These investigators studied children who had experienced accidental finger amputation. They found that the current peaked at 22 microamperes 8 days after the injury and thereafter slowly decreased back to zero. It is believed that this current of injury triggers biologic repair, and later work established that there is actually a battery-like aspect to the epidermis (56-58) that can influence wound
healing. Since membrane potentials are basic in the cell, it is logical to assume that 75 trillion cellular batteries will influence physiology in some way.
Occlusive dressings accelerate wound healing (59). They probably achieve their effects by promoting a moist environment (57) which resurface 40% faster than air-exposed wounds (60). This is possibly related to COI, since a dry wound is less electrically conductive. Electrical stimulation of a wound increases the concentration of growth factor receptors which increases collagen formation (61,62). This may be important in view of the hypothesis that a major mechanism in causing ulceration is removal of growth factors by venous hypertension (63).
Electricity was first used to treat surface wounds over 300 years ago with charged gold leaf to prevent smallpox scars (64). Use of electromagnetic fields predates the application of direct current (54) and there are several studies showing excellent results using this modality (65-68). Animal experiments have shown, however, that direct current can accelerate epithelialization and result in stronger scar tissue formation (69,70).
The first human study using direct electrical current (71) reported complete healing of chronic venous stasis leg ulcers in 3 patients with 6 weeks of treatment. The most frequently cited study (72) used direct currents of 200-1000 microamps in 67 patients. This was repeated in 1976 (73) in 76 patients with 106 ischemic skin ulcers. In 1985 a randomized controlled study was published (74). All of these studies documented significant accelerated healing with electrical stimulation.
In 1974 Rowley et al. (75) studied a group of patients having 250 ischemic ulcers of various types. The series included 14 ulcers in control subjects. The electrically stimulated ulcers had a fourfold acceleration in healing response compared to controls.
A consistent observation in these studies was that wounds that were initially contaminated with Pseudomonas and/or Proteus were usually sterile after several days of electrotherapy. Other investigators have also noticed similar improvement (75-77) and suggest this technique as the preferred treatment for indolent ulcers. No significant adverse effects resulting from electrotherapy have been documented (78) and MET is clearly an effective and safe supplementary treatment for recalcitrant leg ulcers (79). Although most studies use negative current to inhibit bacterial growth and positive current to promote healing, the studies just mentioned used unipolar currents which alternated between positive and negative. There is support for this technique in one animal study (80), suggesting that bipolar current may be better for wound healing (14).
Potential Mechanisms for Repair Stimulation
Becker (49) demonstrated that an electrical current emanating from a biologic control system is the trigger that stimulates healing, growth and regeneration in all living organisms after injury but that this system may become less efficient with time. He theorizes that the self-repair inimical to survival in primitive organisms requires a closed-loop system. A specific injury signal is generated which causes another signal to start
repair. The injury signal gradually decreases over time as the repair process proceeds until it finally ceases when repair is complete. Such a primitive system does not require demonstrable consciousness or intelligence. This purportedly explains why animals actually have a greater capacity for self-healing than do humans.
Becker maintains that it is helpful to compare the nervous system with a digital computer. Both systems transfer information that is represented by the number of pulses per unit of time. Information is also coded according to where the pulses go and whether or not there is more than one channel of pulses feeding into an area. All our senses are based on this type of pulse system, an arrangement similar to that used in computers. It operates remarkably fast and can transfer large amounts of information as digital "off and "on" data.
Becker suggests that early organisms did not need to transmit large amounts of sophisticated information and may have possessed something akin to an analog system which works by means of simple DC currents. This represents information by the strength of the current, its direction of flow, and slow wavelength variations in its strength. Although much slower than the digital model, it is extremely precise and works well for its intended purpose.
Becker theorizes that the first living organisms used this kind of electrical system for injury repair and that we still have this primitive nervous system residing in the perineural cells hidden within the central nervous system. Every nerve cell is surrounded by perineural cells which comprise 90% of the nervous system. They have semiconduction properties which allow them to produce and transmit non-propagating DC signals. This analog system senses injury and controls repair. It controls the activity of body cells by producing specific DC electrical environments in their vicinity. It also appears to be the primary system in the brain, controlling the actions of neurons as they generate and receive nerve impulses.
Although there are concerns that some types of electromagnetic field exposure can cause cancer or leukemia (49,81,82), we have strong evidence that MET can normalize cell growth, accelerate cell division after injury and inhibit cell division when it becomes abnormally accelerated. if a cell is in a normal state of physiologic equilibrium, external electric fields do not appear to affect it (83).
Anti tumor effects of DC currents have been reported (84). The current state of electrical cancer research seems to be where bone repair was about twenty years ago. The only studies published used invasive techniques with percutaneous needle electrodes (8E91). All of the studies report significant impairment of tumor growth with electrical treatment.
Caution is advised during pregnancy because electrical stimulation can affect the endocrine control systems and can theoretically cause miscarriage, although this has never been reported. Microcrrent, or any other electrical stimulus should not be used on patients with demand-type cardiac pacemakers. Other than these two conditions, there are no known significant adverse side effects to MET.
Clearly, much additional work is required to define the role of MET. The results of research published to date strongly suggest that it will have a much more prominent role in the future of health care. In its current form, it can easily and safely control pain and accelerate heatling. Due to its ready availability, cost effectiveness, and safety, it is time for physicians to offer it as an option. The 84% of patients who seek alternative medical techniques would be especially appreciative.
1. Ruoff GE, Beery GB. Chronic pain: Characteristics, assessment, and treatment plans. Postgrad Med 1985; 78:91-97.
2. Eisenberg DM, Kessler RD, Foster C, et al. Unconventional medicine in the United States. Prevalence, costs, and patterns of use. N Engl J Med 1993; 328:246-262.
3. Lake D. Neuromuscular electrical stimulation: An overview and its application in the treatment of sports injuries. Sports Medicine 1992; 13:320-336.
4. Delitto E Snyder-Mackler L. Two theories of muscle stength augmentation using percutaneous electrical stimulation. Physical Therapy 1990; 70:158-164.
5. Leo KC, Dostal WF, Bossen DG, et al. Eflect of transcutaneous electrical nerve stimulation characteristics On clinical pain. Physical Therapy 1986; 6:200-205.
6. Delitto A, Strube MJ, Shulman AD, et al. study of discomfort with electrical stimulation. Physical Therapy 1992; 72:410-424.
7. Melzack R, Wall P. Pain mechanisms: a new theory. Science 1965; 150:971.
8. Bauer W. Electrical treatment of severe head and neck cancer pain. Arch Otolaryngol 1983; 109:382-3.
9. Kirsch D, Lerner F. Innovations In pain management: a practical guide for cilnicians. In: Weiner RL (ed) Electromedicine 1990; Deutsche Press; 23:1-29.
10. Cheng N, Van Hoff H, Bockx E, et al. The effect of electric currents on ATP generation protein synthesis, and membrane transport in rat skin. Clin, Orthop 1982. 171:264-72.
11. Becker RO. The Body ELectric 1985; New York, William Morrow and Co, Inc.
12. Windsor RE, Lester JP, Herring SA. Electrical Stimulation in CIinical Practice. Physician & Sportsmedicine 1993; 21:85-93.
13. Reich JD, Tarjan PP Electrical stimulation of skin. Int J Derm 1990; 29:395-400.
14. Vodovnik L, Karba R. Treatment of chronic wounds by means of electric and electromagnetic fields. A literature review. Med Biol Engineer Comput 1992; 30:257-266.
15. Stanish WD, et al. The use of electricity in ligament and tendon repair. Physician & Sporismedicine 1985; 13:109-116.
16. Stanish Wn. Lai A. New concepts of rehabilitation following anterior cruciate reconstruction. Clin Sports Med 1993; Jan;12(1):25-58.
17. Kulig K, Jarski R, Drewek E, et al. the effect of microcurrent stimulation on CPK and delayed onset muscle soreness. Phys ther 1991; 71:6(suppl).
18. Kaada B, Flatheim E, Woie L. low-frequency tranecutaneous nerve stimulation in mild/moderate hypertension. Clinical Physiology 1991; 11:161-168.
19. North RB, et al. Spinal cord stimulation for chronic, intractable pain: experience over two decades. Neurosurgery 1998; 32:384-395.
20. LeDoux MS, Langford KH. Spinal cord stimulation for the failed back syndrome. Spine 1993; 18:191-4.
21. Neumann V. Electrotherapy. Br J Rheumatol 1993; 32:1-3.
22. Wollersheim H, Van Zwieten PA Treatment of Raynaud's phenomenon. European Heart J 1993; 14:147-49.
23. Mulder, et al. TENS in Raynaud's phenumenon. Angiology 1991; 42:414-17.
24. Engleberg M, Bauer W. Tanscutaneous electrical stimulation for tinnitus. Laryngoscope 1985; 95:1167-73.
25. Shulman A. Subjective idiopathic tinaitus: A unified plan of management. Am J Otolaryng 1992; 13:63-74.
26. Marion MS, Cevette MJ. Tinnitus. Mayo Clin Proc 1991; 66:614-20.
27. Ho RT, Jawan B, Fung ST, et al. Electro-acupuncture and postoperative emesis. Anaesthesia 1990; 45(4):327-9.
28. Crawford PR. Electronic dental anaesthesia. J Can Dent Assoc 1991; 57(6);497-9.
29. Reiss A Electronic dental anaesthesia. Surgery without the needle. Ont Dent 1991;68(10):13-7.
30. Roth PM, Thrash WJ. Effect of transcutaneous electrical nerve stimulation for controlling pain associated with orthodontic tooth movement. A J Orthodontics 1986; 90:132-38.
31. King GE, Jacob RF, Martin JW. Electrotherapy and hyperbaric oxygen: promising treatments for postradiation complications. J Prosthetic Dent 1989; 62:331-334.
32. Barden RM, Sinkora GL. Bone stimulators for fusions and fractures. Orthoped Nursing 1991; 26:89-103.
33. Lavine LS, Grodzinsky AJ. Current concepts review. Electrical stimulation of bone repair. J Bone Joint Surg 1987; 69-A:626-630.
34. Harishorne E. On the causes and treatment of pseudarthrosis and especially that form of it sometimes called supernumerary joint. Am J Med 1841; 1:121-156.
35. Lente RW. Cases of un-united fracture treated by electricity. New York State J Med 1850; 5:317-19.
36. Becker RA, Marino AA. Electromagnetism and Life 1982; State University of New York Press, Albany.
37. Fukada E, Yasuda I. On the piezoelectric effect of bone. J Physiol Soc Japan 1957; 12:1158-62.
38. Bassett CAL, Becker RO. Generation of electrical potentials by bone in response to mechanical stress. Science 1962; 137:1063-1064.
39. Spadaro JA. Electrically stimulated bone growth in animals and man. Review of the literature. Clin Orthop 1977; 122:325-332.
40. Stiller MJ, et al. A portable pulsed electromagnetic field (PEMF) device to enhance healing of recalcitrant venous ulcers: a double-blind, placebo-controlled clinical trial. Br J Dermatol 1992; 127:147-54.
41. Lavine LS, et al. Electric enhancement of bone healing. Science 1972; 175:1118-21.
42. Paterson D. Treatment of nonunion with a constant direct current: A totally implantable system. Orthop Clin North Am 1984; 15:47-59.
43. Brighton CT. The treatment of non-unions with electricity. J Bone Joint Surg (Am) 1981; 63:847-51.
44. Bassett CAL. The development and application of pulsed electromagnetic fields (PEMFs) for ununited fracture and arthrodesis. Orthop Clin North Am 1984; 15: 61-87.
45. Rubin CT, Mcleod KJ, Lanyon LE. Prevention of osteoporosis by pulsed electromagnetic fields. J Bone Joint Surg 1989; 71-A:411-417.
46. Mcleod KJ, Rubin CT. The effect of low-frequency electrical fields on osteogenesis. J Bone Joint Surg 1992; 74A:920-929.
47. Brighton CT, Pollack SR. Treatment of recalcitrant nonunion with a capacitively coupled electric field. J Bone Joint Surg 1985; 67A:577-85.
48. Brighton CT Pollack SR. Treatment of nonunion of the tibia with a capacitively coupled electric field. J Trauma 1984; 24:153-55.
49. Becker RO. Cross Currents 1990; Los Angeles. Jeremy B Tarcher, Inc. 50. Wilson DH. Treatment of soft tissue injuries by pulsed electrical energy. Br S Med 1972; 2:269-70.
51. Binder A, et al. Pulsed electromagnetic field therapy of persistent rotator cuff tendinitis: A double blind controlled assessment. Lancet 1984; I:695.
52. Stanish WD. The use of electricity in ligament and tendon repair. Physician sports Med 1985; 13:108-116.
53. Nessler JP, Mass DP Direct current electrical stimulation of tendon healing in vitro. Clinical Orthopedics 1985; 217:303.
54. Becker RO. The bioelcctric factors in amphibian limb regeneration. J Bone Joint Surg 1961; 43A:643-656.
55. Illingsworth CM, Barker AT. Measurement of electrical currents emerging during the regeneration of amputated fingertips in children. Clin Nays Physiol Meas 1980 1:87-9.
56. Foulds IS, Barker AT. Human skin battery potentials and their possible role in wound healing. Br S Dermatol 1983; 109;515-522.
57. Jaffe LF, Vanable JW. Electric fields and wound healing. Clin Dermatol 1984; 2: 34-44.
58. Barker AT, Jaffe LF; Banable JW. The Glabrous epidermis of cavies contains a powerful battery. Am J Physiol 1982; 242:R358-66.
59. Falanga V. Occlusive wound dressings. Arch Dermatol 1988; 124:872-77.
60. Eaglstein WH, Mertz PM. New methed for assessing epidermal wound healing. The effects of triamcinolone acetonide and polyethylene film occlusion. J Invest Dermatol 1978; 71:382-384.
61. Falanga V, et al. Electrical stimulation increases the expression of fibroblast receptors for transforming growth factor-beta, abstracted, J Invest Dermatol 1987; 88: 488.
62. Alvarez OM, et al. The healing of superficial skin wounds is stimulated by external electrical current. J Invest Dermatol 1983; 81:144-48.
63. Falanga V; Eaglstein WH. The trap hypothesis of venous ulceration. Lancet 1993;341:1006-7.
64. Robinson KR. Digby's receipts. Annals Med History 1925:7:216-19.
65. Jeran M, et al. PEMF stimulation of skin ulcers of venous origin in humans; preliminary report of a double blind study. J Bioelectr 1987: 6:181-88.
66. GoIdin H, et al. The effects of Diapulse on the healing of wounds: a double-blind randomized controlled trial in man. Br J Plast Surg 1981: 34:267-70.
67. Ieran M, et al. Effect of low frequency pulsing electromagnetic fields on skin ulcers of venous origin in humans: a double blind study. J Orthop Res 1990; 8:276-82.
68. Mulder GD. Treatment of open-skin wounds with electric stimulation. Arch Phys Med Rehabil 1991; 72:815-7.
69. Carey, LC, Lepley D. Effect of continuous direct electric current on healing wounds. Surg-Forum 1962; 13:33-35.
70. Assimacopoulos D. Wound healing promotion by the use of negative electric current. Ann Surg 1968; 34:423-31.
71. Assimacopoulos D. Low intensity negative electric current in treatment of ulcers of leg due to chronic venous insufficiency: preliminary report of three cases. Am J Surg 1968; 115:683-687.
72. Wolcott LE, wheeler PC, Hardwicke HM, et al. Accelerated healing of skin ulcers by etectrotherapy. South Med J 1969; 62:795-801.
73. Gault WR, Gatesn PF. Use of low intensity direct current in management of ischemic skin ulcers. Phys Ther 1976; 56:265-69.
74. Carley PJ, Wainapel SF. Electrotherapy for acceleration of wound healing: Low intensity direct current. Arch Phys Med Rehabil 1985; 68:443-446.
75. Rowley BA, McKenna JM, chase GR, Wolcott LE. The influence of electrical current on an infecting microorganism in wounds. Ann NY Acad Sci 1974; 238:543-551.
76. Barron JJ, Jacobson WE. Treatment of decubitus ulcers: a new approach. Minn Med 1985; 68:103-105.
77. Lundeberg TC, Eriksson SV, Malm M. Electrical nerve stimulation improves healing of diabetic ulcers. Ann Plast Surg 1992: Oct;29(4):328-31.
78. Weiss DS, et al. Electrical stimulation and wound healing. Arch Dermatol 1990;126:222-225.
79. Dayton PD, Palladino SJ. Electrical stimulation of cutaneous ulcerations. A literature review. J Am Pod Med Assoc 1989; 79:318-321.
80. Stromberg BV. Effects of electrical currents on wound contraction. Ann Plastic Surg 1988; 21:121-23.
81. Savitz DA, et al. Magnetic field exposure from electric appliances and childhood cancer. Am J Epidemiol 1990; 131:763-73.
82. Pool R. Is there an EMF-cancer connection? science 1991; 249:1096-98.
83. Vodovnik L, et al. Modified cell proliferation due to electrical currents. Med Biol Eng Comput 1992; 30:CE21-28.
84. Humphrey CE, Seal EH. Biophysical approach toward tumor rergression in misc. Science 1959; 130:338-90.
85. Habal MB. Effect of applied DC currents on experimental tumor growth in rats. J Biomed Mater Res 1980; 14:789-801.
86. Nordenstrom BEW. Electrochemical treatment of cancer. Variable response to anodic and cathodic fields. Am J Clin Oncol (CCT) 1989; 12:530-36, 1989.
87. Lyte M, et al. Effects of in vitro electrical stimulation on enhancement and suppression of malignant lymphoma proliferation. J Natl Cancer Inst 1991; 83:116-119.
88. Morris DM, et al. Electrochemical modification of tumor growth in mice. J surg Res 1992; 53:306-309.
89. Sersa G, et al. Anti-tumor effect of electrotherapy alone or in combination with interleukin-2 in mice with sarcoma and melanoma tumors. Anti Cancer Drugs 1992; 3:253-260.
90. Nordenstrom B. Biologically closed electrical circuits: activation of vascular interstitial closed electric circuits for treatment of inoperable cancers. J Bioelectricity 1984; 3:137-53.
91. Belehradek J, Orlowski S, Poddevin B, et al. Electrotherapy of spontaneous mammary tumors in mice. Eur J Cancer 1981; 27:73-76.
Black Box Warnings in Prescription Drug Labeling: Results of a Survey of 206 Drugs Prescription drugs are unavoidably associated with adverse effects, but the ben-efits of using such drugs outweigh the accompanying risks so long as the expert judg-ment of a licensed practitioner is first applied in the decision to use a drug for aparticular patient. Most risks for prescription drugs appear
DESERT DE DANAKIL ET ERTA ALE Ethiopie - B972 Ce voyage aurait pu s'intituler à la naissance du monde ou rendez-vous avec les entrailles de la Terre. Au cœur du désert du Danakil, l'un des plus chauds de la planète, se trouvent deux des plus impressionantes représentations de la vie volcanique de notre planète. Tout d'abord le volcan Erta Ale et son lac de magma de plus de 3000